When you have a patient to whom you need to remove the dental prosthetic (a hybrid on, for example)or whether you have to make a new one or to clean it, you cross your fingers and pray that those definitive screws will follow your fingers ‘movement and will come off. This is very stressful and we all know about it. But we can tell you this: stop worrying!!
If the worst were to happen( loss of screw´s crown), you will be upset the rest of the day…We have worked on a explanatory video that will make an expert of you in this matter, a hero to your patient! Pay attention….
Instructions for the extraction of a dental implant´s screw
Furthermore, you can have the written instructions so the information is clear and you can achieve excellent results
The retriever kit is also useful for dental implants. The kit consists of 4 tips, each one of them for a different indication
Here you have a summary:
RETRIEVER TYPE 1 Ref. ETX ext-1
Retreiver image to retrieve screws with crowned connexion
Designed to retrieve screws with the connexion crowned or deformed. Universal for hexagonals of 1.20, 1.25, 1.27, 1.30, UniGrip, Torx etc. Indicated for screws, scarring pillars etc
RETRIEVER TYPE 2 Ref: ETX-2 ext-2
Designed to retrieve screws with a deteriorated connexion. It is used when the Type 1 retriever does not hold or drags onto the affected area
Screw implants retriever
RETRIEVER TYPE 3 Ref: ETX-3 ext-3
Designed to retrieve screw implants M1.6. Universal for the majority of implants in the market. It must not exceed a torque bigger than 100N-cm. For larger torques, please use retriever type 4.
Retreiver for screw implants
RETRIEVER TYPE 4 Ref: ETX-4 ext-4
Desinged to retrieve screw implants M1.8 and M2. Universal for the majority of implants in the market. It is advisable not to exceed a torque bigger than 200 N-cm. It can also be used in screw implants M1.6 if you have already used the Type 3 retriever.